Proposing Imaging Observations with ACIS
Proposer Threads (Cycle 19)
- ACIS at a Glance
- ACIS Background Issues
- Telemetry Format and Observing Mode
- Choice of Chips in Imaging Mode
These notes provide an overview of the issues to consider when planning a Chandra proposal that uses ACIS. For more comprehensive information, please refer to the references listed below.
The ACIS array consists of a total of 10 CCD chips each with an 8' x 8' field of view on the sky. The layout is shown in the POG Figure 6.2. ACIS-I (the imaging array) consists of a 2x2 array of Front Illuminated Chips and ACIS-S (the spectroscopy array) consists of a 1x6 array of 4 Front Illuminated Chips and 2 Back Illuminated chips. A maximum of 6 chips can be turned on at any given time.
Observers are encouraged to use 4 or fewer CCDs if their science objectives are not significantly affected by turning one or more CCDs off. If the observer's science objectives require 6 CCDs, the observer should set 5 CCDs to "Y" and the sixth CCD to "Off1" in RPS and if the proposal is accepted, the Oberver may work with their User Uplink Support Scientist to change the 6th CCD to "Y". But observers should be aware that 6 CCD observations will be increasingly more difficult to schedule as the mission progresses.
The Chandra Remote Proposal Submission (RPS) software requires proposers to specify which instrument is to be at the aimpoint for a given target. If ACIS-I is selected, the aimpoint is on chip 3. If ACIS-S is selected, the aimpoint is on chip 7. Proposers are also required to submit specific chip selections for each target. In order to prevent the ACIS Power Supply and Mechanism Controller (PSMC) from overheating, the mission planning team may turn off one or more of the chips that the proposer designates as "optional".
If it is necessary to turn chips off due to thermal reasons, the chip designated Off1 will be turned off first, followed by Off2-Off5. Observers should specify the chip set that is best for their primary science. The document "Selecting Required and Optional CCDs" lists and explains in detail recommended chips selections that have proven to be popular, and would facilitate a more useful and homogeneous archive
- Table 6.10 gives background rates for individual chips (in counts/sec/chip) for ACIS-I and ACIS-S at the aimpoint.
The Chandra Remote Proposal Submission (RPS) software requires an estimate of the Total Field Count Rate. This is the sum of all sources in the field of view plus the TOTAL background count rate from all chips that are turned on. Table 6.10 should be used for the total background count rate when calculating the Total Field Count Rate. The numbers in Table 6.10 are much larger than in Table 6.9 because they reflect ALL background counts that are telemetered. Again, the numbers in Table 6.10 are for quiescent periods. Flares will increase the background count rate. The Total Field Count Rate is used to check that the telemetry stream is unlikely to saturate in a proposed observation.
- Faint (F) Format. Gives position, time, and total pulse height for each detected event, plus pixel values in a 3x3 region surrounding the event that characterizes the event grade.
- Very Faint (VF) Format. Same as for Faint, but pixel values in a 5x5 region surrounding the event are telemetered.
- Graded Format. Gives position, time, total pulse height, and grade of the event. Pixel values are not telemetered.
Proposers should be aware that telemetry can saturate in VF observations during a background flare. This possibility can be minimized by utilizing additional filtering on-board (for example, reducing the upper energy cutoff and/or raising the lower energy cutoff) and/or reducing the number of CCDs used in the observation. When applying an energy filter in VF mode, a lower energy of 0.1 and a range of 12.0 is strongly suggested for most observations.
In general, the optimal format is that which telemeters the maximum information without saturation.Two Observing modes are available:
- Timed Exposure (TE) mode. Here the array integrates photons for a fixed period of time (Frame Time) before being read out. The total sky exposure time closely equals the sum of all frames. The result is a two-dimensional image with each event tagged by position, energy etc, depending on the telemetry format chosen. In TE mode, the available Telemetry Formats are Very Faint, Faint, and Graded. It is possible to vary the Frame Time, for example to mitigate pileup (see POG Section 6.16.3).
- Continuous Clocking (CC) mode. In CC mode, rows from the imaging array are continuously clocked into the framestore array, giving a 1x1024 pixel image. One spatial dimension is lost, but the timing resolution is considerably improved (3 msec). In CC mode, the available Telemetry Formats are Faint and Graded. CC mode should only be used for bright sources, and can be useful for mitigating pileup. It should also be noted that the effective background in CC mode is increased by a factor of ~1000!
- If ACIS-I is selected, then the aimpoint is on the Front Illuminated (FI) chip I3. The FI chips exhibit degraded energy resolution near the aimpoint due to Charge Transfer Inefficiency (CTI). This effect can be mitigated somewhat in data processing. The ACIS-I array has a large (16x16 arcminute) field of view.
- If ACIS-S is selected, then the aimpoint is on the Back Illuminated (BI) chip S3. The BI chips are more sensitive at lower energies, and have a more uniform spectral resolution as a function of position on the CCD. The field of view of the ACIS-S array is smaller, and the background in the S3 chip is higher than the FI chips.
- It may be desirable to utilize more restrictive energy filtering than the default and/or to turn off a chip to avoid telemetry saturation in Very Faint mode.
- A restricted region of the chip, or subarray, can be selected to reduce the frametime and provide a faster readout. This can help to avoid telemetry saturation and mitigate pileup.
- The BI chip S1 can be used to monitor flares during an observation and for long-term analysis of background variations. Select to turn on this chip if it isn't detrimental to the science.
- Non-standard chips may be selected, for example to optimally cover an extended object.